There is no doubt that establishing state of the art is a challenge. It ranges from key information on the medical condition to a neutral, comprehensive analysis of treatment options, and culminates in a succinct yet comprehensive presentation that defines the currently accepted safety and performance standards against which the device is measured. Not surprisingly, common pitfalls include a perspective that is too broad and inclusion of too much detail, or a perspective that is too narrow, omitting relevant indications and alternative therapeutic options. An important critical aspect of the clinical evaluation that relies on state of the art data is also to appropriately define the position of the device within the currently available treatment portfolio.(more…)
The large number of mentions of state of the art throughout MEDDEV 2.7/1 rev 4 provide not only a comprehensive description of the importance, purpose, and role of establishing the state of the art, but also inform on how to incorporate this analysis into the clinical evaluation. The image below summarizes several core roles of this analysis.
Prior to MEDDEV 2.7/1 rev 4, guidance referred to state of the art only vaguely. MEDDEV 2.7/1 rev 3 requested the discussion of clinical data “in comparison with” and “taking account of” state of the art, and that the clinical literature data cited “reflect current medical practice and the generally acknowledged state of the art technologies”. From a methodological standpoint, this is not much guidance. Thus, methodology, depth, and presentation of the medical background for a device was largely left to one’s interpretation, and thus conducted inconsistently, or, at least, heterogeneously.